Home Oxygen Concentrator Oxygen Generator
|FOB Price:||Get Latest Price|
|Min. Order:||1 Piece|
|Payment Terms:||L/C, T/T, Paypal, Western Union|
- Model NO.: 1L
- Type: Oxygen Generator
- Application: Home Use
- Trademark: HNC, OEM OR ODM
- Origin: Wuhan, Hubei, China
- Certification: CE, RoHS, ISO 13485, ISO 9001
- Feature: Physical Oxygen
- Flow: 1L/Min
- Specification: CE, ROHS, ISO 13485, ISO 9001
- HS Code: 90189090
1 The oxygen generator is applicable for household use.
2 It is a patented product, and the patent number is zl201020197951. X.
This device generates oxygen purely by physical method, so it contains no chemical substances.
Small in size, it is convenient to use.
The daily cost is no more than 0.12US dollars.
This product applies the streamlined design. The ultra large liquid crystal display is equipped.
The oxygen enrichment accords with the international standard.
6 Highly humid
The generator has high oxygen output with stable oxygen concentration.
1 Relieve the pressure on the brain.
2 Relieve the working pressure
3 home health care oxygen
4 Oxygenating for physical exercise
Oxygen Generation Principle
1 The medical generator can produce 1L oxygen per minute. It is made in accordance with the latest oxygen production technological achievements. As a new high tech product which owns independent intellectual property right, this device has integrated the functions, such as oxygen concentration control, power off alarm, pulverization, air filtering, etc.
2 Firstly, the air enters into the compressor through the inlet silencer. Then it will get into the molecular sieve to separate the oxygen and nitrogen. The oxygen separated gets into the fine screening area, while the nitrogen is absorbed by the molecular sieve and then discharged via the separating valve. After the oxygen is improved in concentration through the fine screening, it will get into the flowmeter which will control the oxygen flow. Next, the oxygen gets into the water tank to be moistened. Finally, it can serve as the supplemental oxygen via the oxygen catheter.
|Voltage/Frequency||220V ± 22V, 50Hz ± 1Hz|
|Oxygen output pressure||0.04-0.06MPa|
|Oxygen flow rate||1L/Min|
|Oxygen concent||≥ 90%|
|Moisture content||≤ 0.07g/m ³|
|Carbon dioxide content||≤ 0.01%|
|Storage conditions||Temperature-40 ° C-55 ° C, humidity< 80%|